BAR Drug Rapid Test
1. Product Details
(1)Product Name: The BAR Drug Rapid Test
2. Direction for Use
Allow the test device, urine specimen and/or controls to reach room temperature (15-30°C) prior to testing.
(1)Bring the pouch to room temperature before opening it. Remove the test device from the sealed pouch and use it as soon as possible.
(2)Place the test device on a clean and level surface. Hold the dropper vertically and transfer 3 full drops of urine (approx. 100ul) to the specimen well (S) of the test device, and then start the timer. Avoid trapping air bubbles in the specimen well (S). See the illustration below.
(3)Wait for the colored line(s) to appear. The result should be read at 5 minutes. Do not interpret the result after 10 minutes.
(1)For healthcare professionals including professionals at point of care sites.
(2)For in vitro diagnostic use only. Do not use after the expiration date.
(3)The test device should remain in the sealed pouch until use.
(4)All specimens should be considered potentially hazardous and handled in the same manner as an infectious agent.
(5) The test device shoud be discarded according to national and local regulations.
The following table lists compounds that are positively detected in urine by the BAR Drug Rapid Test at 5 minutes.
Compound Concentration (ng/mL)
4. Effect of the Urinary pH
The pH of an aliquoted negative urine pool was adjusted to a pH range of 5 to 9 in 1 pH unit increments and spiked with Secobarbital to 150 ng/mL and 450 ng/mL. The spiked, pH-adjusted urine was tested with the BAR One Step Barbiturates Test Device in duplicate. The results demonstrate that varying ranges of pH does not interfere with the performance of the test.
Hangzhou Laihe Biotech Co., Ltd
Address: Room 521 Floor 5 & Floor 1,Building No. 2B,Tianhe High-Tech Park,No. 688,Bin'an Road,Changhe Sub-district,Binjiang District,Hangzhou City,Zhejiang Province,China