BZO Drug Rapid Test
1. Product Description
Name: The BZO Drug Rapid Test
Cut-off: 300 ng/mL(customized)
2. Materials Provided
Disposable specimen droppers
3. Direction for Use
(1)Bring the pouch to room temperature before opening it. Remove the test device from the sealed pouch and use it as soon as possible.
(2)Place the test device on a clean and level surface. Hold the dropper vertically and transfer 3 full drops of urine (approx. 100μl) to the specimen well (S) of the test device, and then start the timer. Avoid trapping air bubbles in the specimen well (S).
(3)Wait for the colored line(s) to appear. The result should be read at 5 minutes. Do not interpret the result after 10 minutes.
NEGATIVE: Two lines appear.
POSITIVE: One colored line appears in the control region (C).
INVALID: Control line fails to appear.
A procedural control is included in the test. A colored line appearing in the control region (C) is considered an internal procedural control. It confirms sufficient specimen volume, adequate membrane wicking and correct procedural technique.Control standards are not supplied with this kit; however it is recommended that positive and negative controls be tested as good laboratory testing practice to confirm the test procedure and to verify proper test performance. Users should follow national and local guidelines for testing QC materials.
Hangzhou Laihe Biotech Co., Ltd
Address: Room 521 Floor 5 & Floor 1,Building No. 2B,Tianhe High-Tech Park,No. 688,Bin'an Road,Changhe Sub-district,Binjiang District,Hangzhou City,Zhejiang Province,China