The MOP One Step Morphine Test Device is a lateral flow chromatographic immunoassay for the detection of morphine s in urine at a cut-off concentration of 300ng/mL. This test will detect other morphine, please refer to analytical specificity table in this package insert.
Drug Of Abuse MOP Morphine Urine Test Cassette
1. Product Description
(1) Product name: Morphine (MOP) Rapid Test Kit
(2) Specification: cassette:40pcs/box, strip: 100pcs/box
(3) Packing: universal packing (foil pouch and paper box)
(4) Cut-off: 300ng/ml
(5) Sample: Urine
(6) Manufacturer: Hangzhou Laihe Biotech Co., Ltd.
(7) Origin: China
(8) Validity: 24 months
(9) Storage: 2-30℃
MOP One Step Morphine Test Strip
MOP One Step Morphine Test Device
2. Materials Provided
(1) Test devices
(2) Disposable specimen droppers
(3) Package insert
3. Materials Required But Not Provided
(1) Specimen collection container
(3) External controls
4. Storage and Stability
Store as packaged in the sealed pouch at 2-30°C.The test device is stable through the expiration date printed on the sealed pouch. The test device must remain in the sealed pouch until use. DO NOT FREEZE. Do not use beyond the expiration date.
5. Directions for Use
Allow the test device, urine specimen and/or controls to reach room temperature (15-30°C) prior to testing.
(1) Bring the pouch to room temperature before opening it.Remove the test device from the sealed pouch and use it as soon as possible.
(2) Place the test device on a clean and level surface. Hold the dropper vertically and transfer 3 full drops of urine (approx.100ml) to the specimen well (S) of the test device, and then start the timer. Avoid trapping air bubbles in the specimen well (S).See the illustration below.
(3) Wait for the colored line(s) to appear.The result should be read at 5 minutes.Do not interpret the result after 10 minutes.
MOP is purifying from Poppy
Hangzhou Laihe Biotech Co., Ltd